Calculates the uncertainty in mass and standard volume flowrate for a liquid orifice meter run. Mass flowrate, density and standard density calculations and their uncertainties are performed using user-selectable calculations based on the density measured using a densitometer. Pressure, temperature, differential pressure and density uncertainties are determined using the technical specification for the transmitters selected from the integrated transmitter selection tool.
The General tab is used to enter information, which is saved along with the uncertainty calculation and included in reports which are generated.
The tabs used for inputting data and calculating the uncertainty in the individual components of the uncertainty can be seen along the top of the module window.
The report tab is only visible once the calculation has been run and will not be visible if any parameters are changed until the calculation has been run once more.
The options section controls how the calculation is performed and dictates which inputs are required.
This option is used to select the number of differential pressure transmitters which are used in parallel. If two transmitters are selected an additional Differential Pressure will be displayed so that information can be entered on both high and low transmitters. If three DP transmitters are selected a third Differential Pressure tab will be included for the mid transmitter.
By default the module will calculate the uncertainty for a single metering stream. If parallel streams are selected the Station Uncertainty will be calculated assuming the metering streams are nominally identical and the flow is split equally between the streams in use. When selecting parallel streams the correlation between inputs has to be considered and the user is presented with a further option to control how the correlation is defined.
Meter run correlation coefficients
If default is selected correlation coefficients which are considered to be typical of this type of meter will be used. Selecting User entered will enable the entry of correlation coefficients in the Flow Rates tab.
This section gives the option of entering the pressure as either an absolute value or as a gauge value with reference to atmospheric (barometric) pressure. If Absolute is selected inputs are required for Absolute pressure and Atmospheric pressure and if Gauge is selected inputs are required for Gauge pressure and Atmospheric pressure. This ensures that whichever is chosen both the gauge and Absolute pressure value is available to other sections of the module; for example the Density calculation tab may require an absolute pressure to perform the NX-19 calculation and the Gauge pressure may be used for the pressure transmitter calculation.
Note: Gauge pressure is equal to the Absolute pressure – the Atmospheric pressure.
This is the differential pressure measured across the orifice plate. If multiple transmitters are used the appropriate transmitter will be selected based on the transmitter ranges and the DP transmitter switch point.
This is the measured temperature of the stream.
This is the line pressure of the stream measured upstream of the orifice plate. As described above, if Absolute is selected inputs are required for Absolute and Atmospheric pressure and if Gauge is selected inputs are required for Gauge and Atmospheric pressure.
The pressure reading on the densitometer.
The temperature reading on the densitometer.
the density of the liquid flowing through the stream.
Number of parallel meters
This input is only visible when parallel streams are selected for the station uncertainty and refers to the number of in-use metering streams the flow is divided between.
The outputs section shows the calculated flow rate and uncertainty taken from the Liquid Orifice plate Flow Rate.
The following tabs are used within this uncertainty module: