Calculates the uncertainty in mass and standard volume flowrates for a liquid flow through a Coriolis meter. Standard density and its uncertainty are manual inputs. Coriolis meter flowrate uncertainty is determined using the technical specification for the Coriolis meter selected from the integrated meter selection tool.
The General tab is used to enter information, which is saved along with the uncertainty calculation and included in reports which are generated.
The tabs used for inputting data and calculating the uncertainty in the individual components of the uncertainty can be seen along the top of the module window.
The report tab is only visible once the calculation has been run and will not be visible if any parameters are changed until the calculation has been run once more.
This section gives the option of entering the pressure as either an absolute value or as a gauge value with reference to atmospheric (barometric) pressure. If Absolute is selected inputs are required for Absolute pressure and Atmospheric pressure and if Gauge is selected inputs are required for Gauge pressure and Atmospheric pressure. This ensures that whichever is chosen both the gauge and Absolute pressure value is available to other sections of the module.
For example: the Density calculation tab may require an absolute pressure to perform an AGA 8 calculation and the Gauge pressure may be used for the pressure transmitter calculation.
Absolute pressure – the Atmospheric pressure = Gauge pressure
This is the line pressure of the gas stream measured upstream of the Coriolis meter.
Note: As described above, if Absolute is selected inputs are required for Absolute and Atmospheric pressure and if Gauge is selected inputs are required for Gauge and Atmospheric pressure.
The default value for Absolute Pressure is set to 1 standard atmosphere (1.01325 bar)
Mass flow rate
This is the mass flow rate through the Coriolis meter.
the density of the gas flowing through the Coriolis meter at standard conditions.
Uncertainty in standard density
the uncertainty in measurement of the density of gas flowing through the Coriolis meter at standard conditions.
This is the measured temperature of the gas stream.
The outputs section shows the calculated flow rate and uncertainty taken from the liquid Coriolis meter tab.
The following blocks are used within this uncertainty module: